Views: 0 Author: Site Editor Publish Time: 2019-09-28 Origin: Site
1. Connect the line of voltage and current back to ensure that the current circuit is not open and the voltage circuit is not short circuit to ensure the normal operation of the device. The device is equipped with a special three-meter stand, and the current conductor on the stand has been connected before delivery. The current output end (red end) on the device is connected to the current input end (on the left) on the stand, and the current input end (black end) on the device is connected to the current output end (on the right) on the stand. The device adopts automatic short-circuit type table holder, when a certain table is not a wall, there is no need to short connect the current loop of the table, which greatly facilitates the calibration of the table.
2. Place the checked table on the table holder and press down the clamping device handle to ensure that the checked table is well connected to the voltage and current loop.
3. Aim the photoelectric head at the turntable and insert the other end into the photoelectric head socket.
4. Turn on the power switch, display interface A, and enter interface B later.5. Power supply of power amplifier for starting device.